Item number: 45.2028
A hygrometer solves this problem for us. This device is based on the property that certain materials have, which is to react to changes in air humidity. In analogue humidity meters, this manifests itself in the form of expansion or contraction. Digital hygrometers integrate components in which conductivity depends on the humidity of the ambient air.
A hygrometer is a device used to determine the relative humidity. Conclusions can be drawn about the quality of the ambient air with the help of the values obtained. Is the air overly dry or too humid? Is there a risk of mould/ mildew in the living spaces? Does my room have the ideal humidity level? Is the humidity level ideal for storing books or musical instruments?
We can answer all these questions based on the hygrometer readings and immediately recognise whether the humidity level is ideal or not.
Today, we have several options to choose from, when it comes to measuring the humidity in the ambient air. Depending on the purpose and preference, you can opt for an analogue or a digital hygrometer.
When combined with a weather station or a WeatherHub Smart Home set, digital hygrometers can benefit from wireless technology which is also used to transmit the measured values from the transmitter to a base station or an app. This way, you can read the hygrometer’s values in a very convenient way and with additional functions such as max-min readings and to set alarm for values.
Analogue hygrometers are also called absorption hygrometers. The measurement is carried out with the help of water-attracting (hygroscopic) materials. The classic amongst analogue humidity meters is the hair-tension hygrometer, which utilises human hair as gauge. In modern hair-tension hygrometers, a synthetic hair is used instead.
The hygrometer was invented by the Swiss physicist Horace-Bénédict de Saussure in the 18th century. He took advantage of the properties of the human hair: when the humidity increases, the hair expands and contracts again as the air becomes drier. Blond hair in particular was considered ideal because it is very thin. At the heart of the TFA-patented and particularly precise synthetic-hair hygrometer lies a synthetic hair. In bimetallic hygrometers, the hair is replaced with a coated bimetallic spring.
TTFA hygrometers are calibrated at the factory. Transport, storage or even permanent use in a dry environment affects the hygroscopic functioning of the bimetallic spring or the synthetic hair.
In order to keep your hygrometer in top shape, we recommend that you regenerate the measuring device at least every six months, calibrate and adjust it if necessary.
Regeneration is primarily about restoring the ideal texture of delicate materials. Hair, for example, can become brittle if constantly exposed to dry air. By exposing the hair to a high-moisture content environment, you restore its natural state, hygroscopic properties and accuracy.
Please proceed as follows:
Please note, however, that a damp cloth does not provide a clear reference value.
If recalibration is required, the hygrometer should be calibrated by comparison against several other hygrometers or a precision reference device (e.g. psychrometer) after regeneration and under normal room conditions.
Please pay attention to the following points:
Adjusting/set analogue hygrometer: instructions
The instructions for regenerating and adjusting analogue hygrometers have been compiled for you as a printable PDF. So you can always keep your device in top shape with the help of our guide.
Additional tips: hygrometer adjusting
The closer the reference value is to the subsequent ambient conditions, the more accurate your measurements will be. In the case of dry rooms, choose a lower spot to adjust it and when using in a humid environment, choose a higher spot instead.
Calibrate or gauge a hygrometer?
In order to determine the exact level of humidity in the air, particularly with analogue hygrometers, care must be taken. We recommend that you regularly regenerate your hygrometer and adjust it if necessary. This process is commonly known as calibration or gauge.
The digital hygrometers from TFA are ready for use after the batteries have been inserted. They measure the current humidity at the installation site within a few minutes after sufficient acclimatization.
Since the quality of the room air always depends on the temperature, we offer combi tools with which you can measure the humidity and temperature simultaneously. TFA’s digital thermo-hygrometers are available in numerous designs and for many special applications.
To determine the humidity perfectly, choose a central, well-ventilated place for the device, not directly next to a heat source or outside wall. The immediate proximity to windows and doors should also be avoided.
If you want to mount your hygrometer to the wall, we recommend that you pay attention to the same conditions. If you have any questions about the best place to measure the humidity, our customer service will be happy to help.
Item number: 30.5020
Item number: 30.3058.01
Item number: 30.5047
+ VARIATIONS+ 360° VIEW
These cookies are essential for the operation of the site and allow, for example, security-related functionality.
|Anbieter||Owner of this website|
|Zweck||Stores the settings of the visitors selected in the Borlabs Cookie Box.|
|Cookie Laufzeit||1 year|
To further improve our website, we collect anonymous data for statistics and analysis. For example, we use these cookies to identify the traffic and the impact of certain pages on our site and to optimize our content.
|Zweck||Cookie from Google for website analysis. Generates statistical data about how the visitor uses the site.|
|Cookie Laufzeit||14 months|
|Zweck||Used to unlock Google Maps content.|
|Cookie Laufzeit||6 months|
|Zweck||Used to unlock YouTube content.|
|Cookie Laufzeit||6 months|